The Shore® Constant Load durometer is designed to minimize the variations in test determinations by the inconsistent application of force when the instrument is used manually.
This durometer incorporates a larger presser foot (51 x 51 mm) for stability and a mass intended to produce a more consistent application of force.
It is designed for testing specimens having minimum dimensions of 51 x 51 mm, the opposite sides of which are flat and parallel. It may be manually applied or, for better repeatability and accuracy, mounted on a Shore® Operating Stand.
Please refer to the Shore® Durometer Selection Guide for an explanation of Shore® model, type and scale designations, as well to assure the proper selection of the Shore® durometer to suit your specific application.
|Types:||A (others by special request)|
|Presser Foot:||51 x 51 mm (2.0 x 2.0 inch)|
|Scale:||0—100, 1 Point Graduations|
|Indentor:||Extension: 2.54 mm (0.100 inch)|
|Standards:||ASTM D2240, ISO 7619 & ISO 868|
Certificate of calibration is included.
|407041000||XACL||A||Constant Load (CL)|
|407041100||XCLMX||A||Constant Load (CL) & Maximum Indicator (MX)|
constant load, half inch foot, standard durometer, quadrant durometer, Shore durometer, Shore durometers, Shore hardness, Shore, Shore A, Shore D, hardness, rubber hardness, plastic, plastic hardness, indentation hardness, ebonite, ebonite hardness, rubber hardness testing, rubber testing, hardness testing, durometer hardness, testing, ASTM D2240, D2240, ASTM D1957, ISO 868, ISO 7619, JIS K6253, JIS K6251, JIS K6252, JIS K 6253, JIS K6254, JIS K6255, JIS K6256, JIS K6257, JIS K6258, JIS K6259, JIS K6260, DIN 53505, DIN 53 505, CNS, K6898, BS 903, BS 2782, BS ISO 21509, BS EN ISO 868, GB/T 10807, Asker, Asker C, SRIS0101, MIL-STD-45622A, MIL-I-45208A, MIL-STD-120, ERDEC, TRDEC, NAVSWC, DREA, 83.060, 83.080.01.